|Home|||||Products|||||Incubator FAQ's|||||News & Events|||||Downloads|||||Contact|||||Links|||||Employment|
The Frequently Asked Questions area is not just for users but for those of you who want to find simple answers to your environmental control needs for live cell imaging. Should your query not be included in the list of Frequently Asked Questions, please E-mail us at firstname.lastname@example.org
No. This type of environmental chamber does not provide precise and accurate temperature control nor does it stabilise focus drift. Both of these parameters are essential for live cell imaging.
I want to use a perfusion chamber. Can you supply one?
No. This type of environmental chamber also does not provide precise and accurate temperature control nor does it stabilise focus drift. Both of these parameters are essential for live cell imaging.
I have made my own chamber. Can I buy just your heater and controller?
In general we do not sell our heaters and controllers separately. Whilst it is possible to adapt our heater and controller for use with your chamber the time taken to physically adapt the heater and then calibrate the device makes such a route beyond economic viability.
Are the Incubation Chambers CE marked?
Yes. Solent Scientific incubation chambers fully comply with the current EC requirements. A Declaration of Conformity is shipped with every incubator.